NEW ProScale Models
Models 190, 290, 390 and 590,
will replace Models 180, 280, 380 and 580 systems. The new systems
use a more robust technology with greater resistance to
electrostatic discharge and electromagnetic interference.
Additionally, the new encoder speed increases to 120 inches
per second. The new models, although not interchangeable with the
older technology, are identical in size and function, making them an
easy, direct replacement for the older systems.
Parts for Models 180, 280, 380
and 580 will remain available for purchase, although complete
systems will no longer be sold.
New systems are compatible with all of the current ProScale Digital
Readouts and accessories
January 1st, 2012 ProTool Database Software has been discontinued.
Important Product Bulletin
Error: “ProCAB-QC Has Stopped Working”
updates for Windows 7® delivered by Microsoft have caused several
installations of ProCAB-QC software to cease to function. When
attempting to start ProCAB-QC, the user will encounter the message:
“ProCAB-QC Has Stopped Working”
The failure between ProCAB-QC and Windows has been traced to
Security update Numbers KB2507938, KB2533623 and
Uninstall these updates.
Until a more
permanent solution is found, you will need to configure your PC’s
Windows Update settings to allow ‘manual selection of updates to
install’ to prevent KB2507938
and KB2533623 from automatically reinstalling.
(Contact your System Administrator or Accurate Technology for
assistance uninstalling updates or configuring update settings)
NEW Firmware Update
A new version of firmware is available for the
following Digital Readout models:
700-1600-235 General Purpose
700-1600-236 General Purpose (w/comp)
700-1600-435 ¼ DIN In-Panel mount
This new firmware addresses a potential issue that may occur when
the batteries in the readout are depleted to a specific low voltage
level AND parameter programming or other updating occurs. In
some cases under these conditions, the memory may be reset to
default factory values without the user’s knowledge.
Accurate Technology, Inc. will provide this firmware upgrade at no
charge. The customer is responsible for shipping charges to our
The occurrence of this issue can be averted by replacing the AA
batteries on a periodic basis every 2 to 3 months for 700-1600-230,
-235, & -236, and 8 to10 months for 700-1600-435, & -700 which
use a 3 volt lithium battery.
If you have any questions, please contact Mark Ehrnsberger in
Engineering at 800-233-0580 or +1 828 654 7920, extension 1315.
Effective May 15, 2010:
GagewaySM RS232 adapter (part # 905-1600-001 ) used to convert
the ProScale Readout SPC data output to RS232 - will no longer
GagewaySM USB adapters will continue to be offered.
Effective Oct. 1, 2009 no new orders for Capacitive Incremental Encoders will be
All warranties for products using these parts will be honored for
the remainder of the warranty.
Products using these parts include:
Capacitive Incremental Encoders look like:
(Click image to enlarge)
These encoders are used with SCALES that look like
Non-Warranty products using this technology will need to be
replaced with the newer Inductive technology (Models 180, 280 & 380). Please call Accurate Technology for additional
information if necessary.